Automatic Capsule Filling Machine integrates machine, electricity and gas. It
adopts microcomputer programmable controller, touch panel operation, frequency
conversion and speed regulation, and is equipped with electronic automatic
counting device. The machine is sensitive in action, accurate filling dose,
novel structure, beautiful appearance, operation Convenient, can automatically
complete the capsule positioning, separation, filling, locking and other
actions, reduce labor intensity, improve production efficiency, and meet
pharmaceutical hygiene requirements. Suitable for filling various domestic or
imported capsules. It is an economical and practical device for filling capsule
drugs in the pharmaceutical industry. As a Pharmaceutical Pharma Bin Supplier,
share with you.
Automatic Capsule Filling Machine
The abnormal situations that occur easily in the operation of the automatic capsule filling machine are as follows:
First, the capsule shell cannot be separated.
1. The capsule shell itself needs to be replaced.
2. The vacuum is too small, you can see by looking at the vacuum meter. The vacuum is too small for many reasons. For the reasons of the vacuum pump itself, ask a mechanic to repair or replace the vacuum pump. The trachea separated from the vacuum pump connected to the capsule shell was broken. Replace the trachea. The filter membrane of the vacuum pump is blocked, and the filter membrane is cleaned. There is also a reason for the small water flow, etc., replace the larger water flow pipe. Specific issues require specific analysis. Here I have only listed a few simple ones.
3. The gap between the module and the capsule shell separator is too large. Either adjust the height of the capsule shell separator, or the module, or the telescopic rod in the module is defective. Analysis is needed to find a solution. Here the medicine powder is too fine or the medicine powder is too sticky, which is prone to problems.
4. The module is not correct. Align the module with the debugging lever until it is automatically seated.
5. The medicine powder is too sticky to block the module. The module should be cleaned.
Twenty-two, the head and tail of the capsule are recessed, and sometimes the capsule shell has a problem. I have encountered a lot of this before, and replacing the capsule shell solved it. Then check whether the height of the thimble needs to be adjusted at the thimble. Sometimes increasing the stress area of the capsule can also be solved. Increasing the stress area of the capsule shell is to customize a thicker thimble. I heard this from the capsule manufacturers, and we didn't do it at the time.
Third, capsule rubbing, this is very headache, and it is often encountered. Need to be treated carefully. There are several reasons for this phenomenon.
1. Capsules and capsules differ too much due to the capsule itself. The capsule shell needs to be replaced.
2. Too much medicine powder remains on the module. The module needs to be cleaned.
3. When the module is installed, the upper and lower molds are not aligned, re-align the module. Until the debugging lever can automatically fall into place.
Our company also has Pharma Bin Series on sale, welcome to consult.